How fast can the 661xC and 663xB Series supplies respond to a command and generate a full scale output?
The output transition time is comprised of two parts: command processing time (before the output starts to change voltage), and the output programming response time (essentially the time it takes the output to go from 10% of the voltage to 90%). Adding both figures will give a good estimate of the output transition time.
For both series of products, the command processing time has been reduced to <4ms.
The 663xB Series products have two operating modes, with a fast/normal mode switch on the rear panel. In the fast mode, the output can slew from 10% to 90% within 400 us. In the normal mode the output slews in 2 ms. Note, use of the fast mode will degrade noise performance somewhat (from 3 mV to 10 mV on the 6632B).
The 661xC models do not offer a fast/normal mode switch, however their output programming response time is approximately 2 ms.
Enroll in myAgilent
A personalized view into the information most relevant to you.
Register Your Product
Register your products for service notifications, firmware update alerts, application notes and more…
Why Buy Agilent
See how the Agilent Advantage adds value to your purchase.
Learn about Keysight
Agilent’s Electronic Measurement Group is becoming Keysight Technologies.