I am utilizing the Agilent 4395A /4396B combination analyzer in spectrum analyzer mode with the noise format. The units are set to V (volts) and the marker indication is 'nV / Hz'. Are the indicated units in error for this mode?
Due to 4395A / 4396B display limitations, the screen annotations are unable to correctly indicate the desired units of 'nV / √Hz'.
The limitations are documented within the related manuals as follows:
Refer to the Agilent 4395A Network/Spectrum/Impedance Analyzer Operation Manual , 'Measuring the Noise Level', page 8-36 and the following excerpt:
Refer to the Agilent 4396B Network/Spectrum/Impedance Analyzer Task Reference, 'Measuring the Noise Level', page 3-3 and the following excerpt:
Enroll in myAgilent
A personalized view into the information most relevant to you.
Register Your Product
Register your products for service notifications, firmware update alerts, application notes and more…
Why Buy Agilent
See how the Agilent Advantage adds value to your purchase.
Learn about Keysight
Agilent’s Electronic Measurement Group is becoming Keysight Technologies.