Is it possible to generate negative voltage using the N6700B Modular Power System?
The output terminals of all N6700B power modules are isolated, and the only limitation is that no output terminal may be more than 240 VDC from any other terminal or from chassis ground. Thus, the positive output terminal of any type of module may be connected to ground (or "common"), which will effectively make that module generate a negative voltage.
The N6784A 4-quadrant General Purpose Source/Measure Unit is a full bipolar power supply that can generate both positive and negative voltages of up to 20 V (at 1 A).
Enroll in myAgilent
A personalized view into the information most relevant to you.
Register Your Product
Register your products for service notifications, firmware update alerts, application notes and more…
Why Buy Agilent
See how the Agilent Advantage adds value to your purchase.
Learn about Keysight
Agilent’s Electronic Measurement Group is becoming Keysight Technologies.