How does averaging of my readings affect the accuracy of the 44702A?
For the 40 mV range:
accuracy = +/- (0.05% + 35 uV + 34 / N^0.5 uV)
where "N" is the number of readings averaged.
For 10 readings averaged this gives +/-(0.05% + 46 uV)
For 100 readings averaged this gives +/-(0.05% + 39 uV)
Similar formulas can be derived for the other ranges using this form and the data book numbers.
Enroll in myAgilent
A personalized view into the information most relevant to you.
Register Your Product
Register your products for service notifications, firmware update alerts, application notes and more…
Why Buy Agilent
See how the Agilent Advantage adds value to your purchase.
Learn about Keysight
Agilent’s Electronic Measurement Group is becoming Keysight Technologies.