How do you trigger a signal using the M9182A and M9183A PXI DMMs?
There are several ways to trigger and measure a signal. Advanced triggering capabilities available with the M9182A/83A PXI DMMs allow you to trigger and measure multiple samples without losing accuracy or resolution (multi-sample mode).
The interactive PDF diagram “Agilent M9182A/M9183A DMM Trigger Operation” provided via link below will guide you. The diagram applies to the Agilent driver (AgM918x.dll) versions 18.104.22.168 and above.
How to use the interactive PDF diagram:
Step 1. As noted at the top of the page, select the trigger model to display by clicking on the green circle next to “Trigger Model” or “Multi-sample model”.
Step 2. On the right-hand side of the diagram, to display the relevant information, click to check applicable green cell.
Step 3. On the left-hand side of the diagram within each detail box, click on the arrows “>” in the upper right corners to reveal additional information.
Documents & Downloads
Agilent M9182A/M9183A DMM Trigger Operation
Misc Item 2011-08-01
PDF 1.25 MB