What is the amplitude resolution of the marker readout in the 8560E and 8560EC series spectrum analyzers?
Although a 12-bit digitizer is used in the 8560E and 8560EC spectrum analyzers, the ADC alone does not determine the Y-axis or vertical resolution of the display. Design considerations for overall display resolution are also driven by the X-axis resolution, which is 601 data points or "buckets" for this family of analyzers. As a result, some ADC vertical resolution is discarded.
In log scale, the resolution is:
|5 dB/div||1/12 dB|
|2 dB/div||1/30 dB|
|1 dB/div||1/60 dB|
In linear scale, the resolution is:
- Reference Level / 600 dB
There is no way to extract the digitized data from the analyzer immediately after the ADC.
Enroll in myAgilent
A personalized view into the information most relevant to you.
Register Your Product
Register your products for service notifications, firmware update alerts, application notes and more…
Why Buy Agilent
See how the Agilent Advantage adds value to your purchase.
Learn about Keysight
Agilent’s Electronic Measurement Group is becoming Keysight Technologies.