Can U1602A/B and U1604A/B measure capacitance of 0.13nF?
Based on the datasheet, for 0.13nF, it falls in the range of 60nF. To calculate the range of fluctuation of the reading, it is given to be 2% of reading and 0.2% of range. So, the range of error is 0.2% x 60nF = 0.12nF. This is already close the 0.13nF capacitance that you would like to measure. Therefore, it is unsuitable to be used to measure such a small capacitance.
Enroll in myAgilent
A personalized view into the information most relevant to you.
Register Your Product
Register your products for service notifications, firmware update alerts, application notes and more…
Why Buy Agilent
See how the Agilent Advantage adds value to your purchase.
Learn about Keysight
Agilent’s Electronic Measurement Group is becoming Keysight Technologies.