On the ENA 6 series, can I get TTL signal indicating Pass/Fail of the limit test like the 871x?
(SUPPORTED MODELS IN BOLD: E5071C E5070B/71B E5070A/71A E5061A/62A E5061B E5072A)
Yes. The Handler interface can be used. Note that the definition is different from the 871x. In the E5061A/62A and E5061B, level-high means "Pass" and low means "Fail".
Enroll in myAgilent
A personalized view into the information most relevant to you.
Register Your Product
Register your products for service notifications, firmware update alerts, application notes and more…
Why Buy Agilent
See how the Agilent Advantage adds value to your purchase.
Learn about Keysight
Agilent’s Electronic Measurement Group is becoming Keysight Technologies.