When the frequency-offset is turned on, the number of sweeps during measurement increases. Is this the normal operation for the ENA?
(SUPPORTED MODELS IN BOLD: E5071C E5070B/71B E5070A/71A E5061A/62A E5061B E5072A)
Yes. If the frequency-offset function is turned ON, the number of sweeps during measurement will increase. For example, the ENA needs two sweeps to measure the S-parameters of a full two-port calibrated setup in a normal frequency sweep; however, four sweeps are required when the frequency-offset function is turned ON. This holds true regardless of whether the stimulus and the receiver ports have the same frequency setting.
Enroll in myAgilent
A personalized view into the information most relevant to you.
Register Your Product
Register your products for service notifications, firmware update alerts, application notes and more…
Why Buy Agilent
See how the Agilent Advantage adds value to your purchase.
Learn about Keysight
Agilent’s Electronic Measurement Group is becoming Keysight Technologies.