The red "Failed" LED comes on right away and stays on. Is there something wrong?
The HP E6432A is a register-based instrument. Part of its self-test requires established communication with the plug&play driver residing in the host controller. The driver must run an initialization routine to finish self-test. After this, the red LED should extinguish. If it comes on during normal operation, it indicates a hardware failure.
Enroll in myAgilent
A personalized view into the information most relevant to you.
Register Your Product
Register your products for service notifications, firmware update alerts, application notes and more…
Why Buy Agilent
See how the Agilent Advantage adds value to your purchase.
Learn about Keysight
Agilent’s Electronic Measurement Group is becoming Keysight Technologies.