Why is there a notch in the noise floor of the 8593EM & 8596EM EMC analyzers at 6.5 GHz when making a radiated prescan measurement?
This is one of the crossover points for the different bands of the instrument where the internal circuitry switches into the next frequency range.
Enroll in myAgilent
A personalized view into the information most relevant to you.
Register Your Product
Register your products for service notifications, firmware update alerts, application notes and more…
Why Buy Agilent
See how the Agilent Advantage adds value to your purchase.
Learn about Keysight
Agilent’s Electronic Measurement Group is becoming Keysight Technologies.