When programming the 3852, what can I do when I get the error "Error 51 -- Symbol Table Full"?
The size of the symbol table at power-up allows 138 entries. These entries represent all declared and downloaded variables, arrays, and subroutines.
If the error message above is received, the size of the symbol table can be increased using the SYMSIZE command. This command requires that the mainframe have revision 3.0 or greater firmware. The Command Reference manual discusses the use of this command further.
Note: The symbol table size is stored in volatile memory and is therefore reset to the default value after RESET and power cycling.
Enroll in myAgilent
A personalized view into the information most relevant to you.
Register Your Product
Register your products for service notifications, firmware update alerts, application notes and more…
Why Buy Agilent
See how the Agilent Advantage adds value to your purchase.
Learn about Keysight
Agilent’s Electronic Measurement Group is becoming Keysight Technologies.