Is there a battery in the Agilent 34970A?
Yes. A 3.0-volt lithium battery, A1BT101, is used to power the real-time clock, stored state, and reading storage memory with line power removed. The battery has a nominal lifetime of 4 years and can be replaced by the user. The battery is soldered into the 34970A. This battery does not affect the calibration memory.
A low battery will typically give errors when ac line power is removed and then re-applied. Any of the following errors may indicate a low battery:
- 201, "Memory lost: stored state"
- 202, "Memory lost: power-on state"
- 203, "Memory lost: stored readings"
- 204, "Memory lost: time and date"
Replacement battery Agilent part number 1420-0860. Refer to the 34970A Service Guide for verification and replacement procedures.
Enroll in myAgilent
A personalized view into the information most relevant to you.
Register Your Product
Register your products for service notifications, firmware update alerts, application notes and more…
Why Buy Agilent
See how the Agilent Advantage adds value to your purchase.
Learn about Keysight
Agilent’s Electronic Measurement Group is becoming Keysight Technologies.