Why does a power meter show the CAL OUT power to be 2-3dB higher than specified on the 8590 Series Spectrum Analyzer?
The front panel CAL OUT signal contains a significant harmonic content. The power meter will read a higher power due to the harmonics of the calibrator. This may be eliminated by using a low pass filter. The correct measurement procedure is given in the 8590 Series Analyzers Calibration Guide, part number 08594-90183
Enroll in myAgilent
A personalized view into the information most relevant to you.
Register Your Product
Register your products for service notifications, firmware update alerts, application notes and more…
Why Buy Agilent
See how the Agilent Advantage adds value to your purchase.
Learn about Keysight
Agilent’s Electronic Measurement Group is becoming Keysight Technologies.