How do I make spectrum analysis measurements on signals above 110 GHz using a PSA Series Spectrum Analyzer (E4440A, E4446A, E4448A)? Do I need additional instruments or accessories?
The PSA with option AYZ enables the use of external mixers. Mixers above 110 GHz use the same port for LO in and IF out. In order to connect these types of mixers to the PSA, a diplexer is required. The purpose of the diplexer is to split the IF out so it can be connected to the IF port and combine the LO on to a single line connected to the mixers LO/IF line. A harmonic of the LO from the PSA mixes with the incoming RF and an IF signal is generated. The IF is low pass filtered in the diplexer and sent to the IF port of the PSA.
External harmonic mixers above 110 GHz generally require biasing. The PSA supplies bias current in the range of +/- 10 ma. This bias current is supplied on the IF port.
Enroll in myAgilent
A personalized view into the information most relevant to you.
Register Your Product
Register your products for service notifications, firmware update alerts, application notes and more…
Why Buy Agilent
See how the Agilent Advantage adds value to your purchase.
Learn about Keysight
Agilent’s Electronic Measurement Group is becoming Keysight Technologies.