Can I use a 50 ohm network analyzer to measure 75 ohm devices?
This is generally not recommended since a 50 ohm test set is optimized to measure at 50 ohms. Measuring 75 ohm devices on a 50 ohm test set is not as accurate as measuring on a 75 ohm test set. However, it can be done. Here are the procedures and the caveats.
Attach a 50 to 75 ohm pad and calibrate at the 75 ohm plane with a 75 ohm calibration kit with the system impedance set at 75 ohms. The raw directivity is degraded by twice the value of the pad in dB.
Use a 50 to 75 ohm mechanical adapter. Again, calibrate at the 75 ohm plane with a 75 ohm cal kit with the system impedance set at 75 ohms. Mismatch effects will affect the accuracy of the calibration and measurement. The Agilent model 11852B minimum loss pad provides a 50 to 75 ohm impedance matching function from DC to 3GHz. The standard model has a Type-N(f) 50 ohm connector and a Type-N(m) 75 ohm connector. Option 004 provides a Type-N(m) 50 ohm connector, and a Type-N(f) 75 ohm connector.
Revision 2.0 25 January 2000
Enroll in myAgilent
A personalized view into the information most relevant to you.
Register Your Product
Register your products for service notifications, firmware update alerts, application notes and more…
Why Buy Agilent
See how the Agilent Advantage adds value to your purchase.
Learn about Keysight
Agilent’s Electronic Measurement Group is becoming Keysight Technologies.