Why does my ESG report errors when I try to download waveforms using a National Instruments GPIB card with certain Signal Studio applications?
If errors reported by the ESG when trying to download waveforms using a NI GPIB card with a Signal Studio application in this scenario include “ -113 undefined header” and “-101 invalid character”, you need to download and install a patch for the Agilent IO Libraries (Suites 14.1 or 14.2 only). The file below includes the patch and instructions for installing the patch
Enroll in myAgilent
A personalized view into the information most relevant to you.
Register Your Product
Register your products for service notifications, firmware update alerts, application notes and more…
Why Buy Agilent
See how the Agilent Advantage adds value to your purchase.
Learn about Keysight
Agilent’s Electronic Measurement Group is becoming Keysight Technologies.