E4916A Crystal Impedance Meter / LCR Meter Firmware Update
Caution: Please contact to near Agilent service center if it is required to upgrade FW of E4915A/E4916A. Do not operate these steps yourself.
Download Instructions for Firmware
- Click on the " E4915A/E4916A Firmware Revision X.XX" listed below the page. Copy the file( .m ) to a floppy disk.
- Setup the PC environment which is used for adjustment and peformance tests of E4915A/E4916A. See Adjustment and Performance tests chapter in E4915A/E4916A Service Manual
- Connect the E4915A/E4916A to the PC through the GPIB cable.
- Start the Adjustment/Performance Test Program.
- Select "INSTALL_FW" in the list box on the program display, then click "OK".
- Follow the instruction of the program.
See the Firmware Installation Procedure in the E4915A/E4916A Service Manual.
|Release Date||Version||Version Description|
|2001-07-01||2.13||E4915A/E4916A Firmware Revision 2.13|
How to Download
- Click the download button and save the file to a temporary directory on your PC (or instrument).
How to Install
- Does Not Require a License
- No specific prerequisites
- E4915A E4916A
Enroll in myAgilent
A personalized view into the information most relevant to you.
Register Your Product
Register your products for service notifications, firmware update alerts, application notes and more…
Why Buy Agilent
See how the Agilent Advantage adds value to your purchase.
Learn about Keysight
Agilent’s Electronic Measurement Group is becoming Keysight Technologies.