Firmware for U1610A/U1620A Handheld Digital Oscilloscope
Always update your handheld oscilloscope with the latest firmware to maximize the performance of the instrument. Please follow all the instructions below step-by-step to ensure the firmware installation is successful through USB drive.
Ensure there are sufficient power supply and no power interruption to the handheld scope during the firmware loading process.
It’s recommended to connect the power source to the handheld scope during the firmware loading process.
Pre-requisite: Ensure the handheld scope is loaded with firmware version 130.4 or above before proceed to version 150.0. You are required to upgrade the handheld scope to version 130.4 if the current version is 126.1 / 128.21 / 129.0
Follow these steps to upgrade the firmware:
NOTE: The firmware upgrade will reset system settings and erase waveform setup. Only waveform traces can be retrieved in the new firmware version.
Step 1: Download the firmware.zip file from the website, unzip and copy the files “updateNewATN.bin” & “updateNewATO.bin” to a clean USB thumb drive (in root directory).
Step 2: Plug the USB thumb drive into the handheld scope.
Step 3: Press “User” button and select “Host” by toggling (pressing F1) the selection under the USB menu.
Step 4: Press “User” button and select “Service” by pressing F4 key.
Step 5: Select “Update firmware” by pressing F1 key.
Step 6: Choose OK to proceed with the firmware update.
Step 7: The unit will update the firmware and reboot automatically.
Do note that system will takes approximately 1 minute for the copying process to take place (DO NOT press “OK” repeatedly during this process).
Step 8: The unit will update the firmware and reboot automatically.
Step 9: Press “User” and select “System Information” (F5) key to check the firmware revision as follows:-
- Firmware: 0.150.0.C (For 100MHz scope, U1610A)
- Firmware : 0.150.0.A (For 200MHz scope, U1620A)
NOTE: Firmware update files “updateNewATN.bin” and “updateNewATO.bin” can only be used for one time update only. Users need to delete existing and copy a new set of files to a clean USB thumb drive if they want to upgrade for others handheld unit.
|Release Date||Version||Version Description|
|2013-02-22||0.150.0||Firmware Installation Instructions for U1610A/U1620A Handheld Digital Oscilloscope - (Rev 0.150.0)|
How to Download
- See Above
How to Install
- See Above
- Does Not Require a License
Add-On Software / Utilities
- Windows 7
- Windows Vista
- See Above